Schick 33/Elite & Fona Stellaris & XIOS Select / Supreme Cable Replacement Movie
IMPORTANT! - Like other electronic devices, your digital x-ray sensor is susceptible to electrostatic discharge (ESD), particularly when the device is used in or around carpeted areas or low-humidity environments. During cable replacement, when Sensor contacts are exposed, it is especially important to protect the device from potential ESD damage. Touching a metal surface prior to replacing the Schick Elite/Schick33/ Fona Stellaris cable will reduce the risk of damaging Sensor components by accidental static discharge. The use of anti-static floor mats or floor treatments (for example Staticide 2005/2002) will also help eliminate static build-up in your office. Make sure the Sensor is placed securely on a clean, moisture-free surface.
Please clean and dry your hands before performing this procedure. Do not wear gloves since the powder inside them could be deposited on sensor contacts while replacing the cable.